*New: Control: First Release - HTA new models 2000H, 2000HT, 2100H could be controled by the HT2x00H driver after selecting corresponding protocol version on the sampler.
*New: Control: First Release - QuadroCore MCore control module is now available in version 126.96.36.199.
*New: Control: First Release - Shanghai Sunny Hengping Scientific Instrument GC-MS control module is now available in version 188.8.131.52.
*New: Control: Update - Apix ChromPix2 control module has been updated to version 184.108.40.2061.
*New: Control: Update - Ecom ECP2000 control module has been updated to version 220.127.116.11.
*New: Control: Update - Ecom TOY control module has been updated to version 18.104.22.168.
*New: Control: Update - Futecs CT-6000 control module has been updated.
*New: Control: Update - Sykam S5300 control module has been updated to version 22.214.171.124.
*New: Control: Update - Young In Chromass YCChroZenAS control module has been updated to version 126.96.36.199.
*New: Control: Update - Young In Chromass YCChroZenGC control module has been updated to version 188.8.131.52.
*New: Control: Update - Young In Chromass YCChroZenPump control module has been updated to version 184.108.40.206.
*New: Control: Update - Young In Chromass YL6500 GC control module has been updated to version 220.127.116.11.
*New: Control: Update - Young In Chromass YL9110 control module has been updated to version 18.104.22.168.
*New: Control: Dani GC1000, Master GC, Master AS, Master DHS, Master SHS control modules are moved from Typical to Full installation.
*New: Control: Fraction Collector by Multiposition Valve - default vial range set to positions 2-n (position 1 assigned to Waste).
*New: Control: Fraction Collector by Multiposition valve - new option to reset position to Start Vial after defined number of fractions.
*New: Control: Fraction Collectors - switching to waste between lines avoided in case of consecutive fractions (Stop time = Start time on next line) with signal condition None.
*New: Control: Young In Chromass - YCChroZen Pump comment is changed from YI9619 Binary Pump to YI9613 Binary Pump and YCChroZen HPLC-E comment from YL9675 RI detector to YI9673 RI detector.
*New: Audit Trail: Improved logging of pop-up message boxes.
*New: Audit trail: Improved user identification in system-generated messages.
*New: Batch: Improved user interface to ease up working with the dialog.
*New: Clarity: Improved computer memory management while working with larger method files.
*New: Clarity: New graphic style of splashscreen and About dialog.
*New: Clarity: New model of licensing system is introduced containing paid major upgrades. More details can be found at www.dataapex.com/upgrade.
*New: Clarity: The appearance of windows now reflects the appearance of Windows 11.
*New: Communication recording: New option TextMode=BOTH (both text and hexa format) in commdrv.ini.
*New: Export to Excel: Implemented support both for old *.xls and new *.xlsx format. Excel version supporting the selected format must be installed on the computer.
*New: Export: While exporting sequence, Sample type is now exported as it is visible in the software, not as a number (0-3), which referred to the order in the menu.
*New: File Open dialogue: Column width is now stored in the user desktop (*.dsk) file.
*New: Instrument Image: Custom Instrument images for multiple chromatographs were added.
*New: Integration Algorithm: Current Integration Algorithm is renamed to Legacy and updated to version 9.0 rev. 1.
*New: Integration Algorithm: A new integration algorithm Wave based on previous Experimental algorithm is added.
*New: Integration Algorithm: Selection of Integration Algorithm is moved to Integration tab (originally on Calculation tab in Method Setup and Results tab in Chromatogram).
*New: Method: Warning when method size may exceed 200 MB during an attempt to save it is now present to help prevent exceeding the memory.
*New: Sequence: New option "On Enter - Move Down" in User Options to handle Enter in Sequence table (useful for entering values using handheld barcode scanners).
*New: SST: Amended selection for performance calculations reflecting the change in USP since 12/2022.
*New: System Configuration: Warning after update that some configured devices will be removed because their driver is not installed is now present.
#Fixed Bug: Control: Agilent HP7673 - Reopening the instrument without a Clarity restart could result in a communication error.
#Fixed Bug: Control: Agilent ICF - It was not possible to uninstall Agilent ICF drivers via Uninstall.exe.
#Fixed Bug: Control: Agilent ICF DAD detectors - spectral detector not created on Windows in Japanese and Portugal localization.
#Fixed Bug: Control: Agilent ICF GC control - Possible deadlock when configuring unconnected GC.
#Fixed Bug: Control: Agilent ICF - Time Idle event performed immediately after finishing the previous run instead of in a given time under some circumstances.
#Fixed Bug: Control: Dani GC1000 - some parts of method were not sent (introduced in version 8.6.0).
#Fixed Bug: Control: Fraction Collector by Multiposition Valve - Wrong behavior when using the Collect All option.
#Fixed Bug: Control: Fraction collector by Multiposition Valve - Wrong reaction to Next or Go to Position before first Collecting.
#Fixed Bug: Control: Fraction Collector by Multiposition Valve - Wrong behavior in Device Monitor on last Vial position.
#Fixed Bug: Control: Recipe EC6000 - The start when using external autozero contact was not functional in a Single Analysis mode when method was not resent.
#Fixed Bug: Control: Shimadzu GC-2010 - The device monitor was not displayed correctly when increased display scaling was used.
#Fixed Bug: Control: Spark Endurance - fixed problems with method editing and printing.
#Fixed Bug: Control: Sykam S2100 - Warning about exceeding Max Pressure limit in LC Gradient options was missing.
#Fixed Bug: Control: Sykam S5200 crash in device monitor in other than English localization.
#Fixed Bug: Analysis: File write error while creating chromatogram in case invalid characters (ASCII 0-31) were used in Sample or Sample ID and propagated to the chromatogram file name.
#Fixed Bug: Analysis: Single Analysis using sampler stayed in Waiting for injection in case of autostop in the method was not enabled.
#Fixed Bug: Audit Trail: Export of file-specific (Method etc.) audit trails were not logged to Station Audit Trail.
#Fixed Bug: Batch: During Batch sequence reprocessing, the options for calibration cloning were ignored.
#Fixed Bug: Calibration: Opening calibration with a high number of signals, compounds, or concentration levels was slow.
#Fixed Bug: Calibration: Wrong handling of customized units for the area - affecting linearised response and calibration curve graph.
#Fixed Bug: Data Acquisition: Wrong synchronization of auxiliary signals start under certain circumstances.
#Fixed Bug: Export Data - option Chromatogram: The signal was exported with an incorrect unit label in some cases.
#Fixed Bug: Export: Export Chromatogram did not respect settings in User Options - Directories.
#Fixed Bug: File Open Dialog: Pressing Esc didn't close the file open dialog when none of the files was selected.
#Fixed Bug: File Open dialogue: Fixed problems with All Signals checkbox focusing when selecting signals of a chromatogram.
#Fixed Bug: GPC: Coefficients for the polynomial calibration curve were displayed with 5 decimals (instead of 4 valid digits, resulting in rounding of small numbers).
#Fixed Bug: High DPI displays: Icons and table columns were too small when higher DPI was used.
#Fixed Bug: Integration Algorithm: Tangent skimming method and Moving Average filter - incorrect evaluation in some cases.
#Fixed Bug: MS Search: MS search in NIST libraries from Clarity used the wrong spectral range.
#Fixed Bug: PDA: When using Add signal from the PDA window Global Bunching was not applied.
#Fixed Bug: Print to PDF: Lab Footer was not printed to PDF when using 1600*1200 display resolution.
#Fixed Bug: SST: Possible crash after creating chromatogram with SST method.
#Fixed Bug: Summary table: Chromatogram was not displayed if there was no peak with the calculated amount. Now sufficient conditions for the chromatograms to be displayed are connected calibration files.
#Fixed Bug: Summary Table: Zeros instead of ??? wrongly displayed in user columns in case used variables were not available.
#Fixed Bug: System Directories: Problems with finding file locations in case the mapped disk notation was used to define the datafiles path.